Class I Recall of Physio-Control Recalls LIFEPAK15 Monitor and Defibrillator
Physio-Control is recalling the LIFEPAK Monitor/Defibrillator due to an electrical problem which may prevent the device from delivering the electrical shock needed to revive a patient in cardiac arrest. If the electrical shock is not delivered, the monitor will indicate “Abnormal Energy Delivery” on the display. A delay in delivering this therapy could result in serious patient injury such as permanent organ damage, brain injury, or death.
- The LIFEPAK 15 Monitor/Defibrillator (LP15)
- Serial Numbers: See full list in recall notice, link below
- Distribution Dates: September 13, 2014, to July 4, 2016
- Manufacturing Dates: September 3, 2014 to June 30, 2016
- Devices Recalled in the U.S.: 139 nationwide
On January 12, 2017, Physio-Control sent an “Urgent Medical Device Correction” letter to all affected customers. The letter asked customers to:
- Review the safety notice and ensure appropriate staff is aware of the notice.
- Follow the instructions on the “Confirmation Sheet” for each serial number listed in the facility’s possession, and return the completed Confirmation Sheet to Physio-Control.
- Perform the “User Test” every day, as instructed in the LIFEPAK 15 Operating Instructions, page 10-4. If the device fails the “User Test”, as described in the notice, contact Physio-Control immediately to arrange for correction of the device.
- Continue to perform the User Test as outlined in the Daily Operator’s Checklist within the LIFEPAK 15 Operating Instructions.
See the recall notice for additional information.
Want to read the full alert and receive alert emails?