FDA Alert – Potential Risks with Fluid-filled Intragastric Balloons
The US Food and Drug Administration (FDA) has recently received multiple reports for two different types of adverse events associated with fluid-filled intragastric balloons used to treat obesity. The first type of adverse event involves the fluid-filled intragastric balloon over-inflating with air or with more fluid (spontaneous hyperinflation) in patients’ stomachs, resulting in the need for premature device removal. The second type of adverse event is the development of acute pancreatitis, which has also resulted in the need for premature device removal.
The FDA wants health care providers to be aware that these potential complications have been observed in patients treated with FDA-approved obesity treatment devices, although the root cause and incidence of these complications is not known at this time.
Intragastric balloon systems are weight-loss systems to treat obesity, which function by taking up space in a patient’s stomach. In 2015, the FDA approved two intragastric balloon systems in the U.S. The ReShape Integrated Dual Balloon System is manufactured by ReShape Medical Inc., and uses two balloons. The Orbera Intragastric Balloon System manufactured by Apollo Endo-Surgery uses one balloon. Both brands are fluid-filled balloon systems, and are indicated for use in conjunction with diet and exercise. The balloons are placed into the stomach through the mouth, using a minimally invasive endoscopic procedure, while the patient is under mild sedation. Once in place, the Orbera balloon is filled with saline only, and the ReShape balloons are filled with saline and methylene blue dye. The maximum placement period for both brands is six months.
In 2016, the FDA approved the Obalon Balloon System, which consists of up to three balloons that are delivered to the stomach inside a swallowable capsule. The capsule is attached to a thin inflation catheter used to inflate the balloons with air. To date, the concerns addressed in this letter have only been observed with the use of fluid-filled intragastric balloons and have not been reported for the Obalon Balloon System.
Over-Inflation – The FDA has received several dozen adverse event reports regarding balloon over-inflation of fluid-filled systems. Most have involved the Orbera Balloon System although this type of event has also been reported with the ReShape Balloon System. Symptoms of balloon over-inflation included intense abdominal pain, swelling of the abdomen (abdominal distension) with or without discomfort, difficulty breathing, and/or vomiting. The event may require premature removal of the balloon to resolve the patient’s symptoms. The reports indicate that balloon over-inflation can occur as soon as nine days after implantation. As the current labeling for fluid-filled intragastric balloons does not address the issue of spontaneous over-inflation, health care providers, including ER professionals, may not be aware that the symptoms listed above could potentially be related to an over-inflated balloon. At this moment there is not enough information to determine what is causing the balloon to over-inflate.
Acute Pancreatitis – A separate set of reports involving both brands of fluid-filled balloons (Orbera and ReShape) indicate that acute pancreatitis developed in several patients due to the compression of gastrointestinal structures created by the implanted balloon(s). All of these cases required premature device removal, and four of the patients required hospitalization. Symptoms of acute pancreatitis in the presence of a fluid-filled intragastric balloon system included severe abdominal and back pain. The reports indicated that pancreatitis can occur as soon as three days after implantation. Pancreatitis is not listed as a potential complication in the labeling information for the Orbera or for the ReShape balloon systems. Therefore, health care providers, including ER professionals, may be unaware that acute pancreatitis could develop following balloon placement. This could lead to unsuitable or delayed treatment for pancreatitis in patients with balloons who may have acute abdominal pain.
The FDA recommends that health care providers:
Recognize that patients with implanted fluid-filled intragastric balloons may develop balloon-related symptoms or other abnormalities following balloon placement, and throughout the duration of their treatment. Consider spontaneous over-inflation and/or pancreatitis in the differential diagnosis of patients presenting with the symptoms noted in this communication. If abnormalities are found, perform any confirmatory diagnostic studies. If the device is removed, follow the manufacturer’s instructions for device returns or evaluations.
Report any adverse events related to intragastric balloon systems that come to your attention. Voluntary reports can be submitted through FDA’s MedWatch Reporting System.
Follow the link below for additional information.
Want to read the full alert and receive alert emails?